"Being in the foxhole together is not just a partnership; it's the essence of collaboration. For it to work, dentists and technicians must truly understand each other's challenges and goals. That's when you see the best possible results and the most satisfied patients." – Terry Fohey, CDT
LMT Original Research © 2010 - 2015
Anyone have any advice on zirconia bridge frameworks "warping"? Hardly ever happens with frameworks with no curve, but seems to be more frequent with 4 unit and larger frames that curve (anterior). Individual units fit terrific internally and marginally, but 2-3 out of 10 times the frames will rock slightly....
The most common cause of that for me would be weak or improperly positioned connectors. The zirconia likes to be uniform and shrink as a whole. Thin areas tend to shrink faster (for obvious reasons). Connectors that are too far lingual or facial also seem to "bend" the frame slightly due to the thicker, less evenly distributed portions on either side.
84 South Main St., Newtown CT 06470
203-426-4LMT · 203-491-2300 (fax)
LMT Communications delivers the business strategies, information, dental laboratory news and education that dental laboratory decision-makers need to succeed. Through LMT Magazine and our LAB DAY shows, LMT works to advance excellence in dental technology, enhance the products, services and communication of dental technicians, and give dental laboratory owners and managers the tools they need to flourish. Learn More »